Please login if you have access to particular applications.
|Subject:||Fixed Butanol restriction|
|When:|| 2017-09-03 08:09:23 UTC |
- entered into Elog/Everlog as 2017-09-03 08:09:23
Condensation Particle Counter (TSI 3772) [view]
|Position:||Latitude: -27.1851 Longitude: 155.9227 Position from underway logging|
The butanol inlet tube was too long, and butanol was not getting in to the chamber. I shortened the tube and it appears to be working