Please login if you have access to particular applications.
|When:|| 2018-01-14 05:39:39 UTC |
- entered into Elog/Everlog as 2018-01-14 05:39:39
|Equipment:|| Clid radar
Bistatic Radar System for Atmospheric Studies (BASTA) cloud radar [view]
Latitude: -44.01317 Longitude: 147.13267
- entered into Elog as 44 00.79 S, 147 07.96 E
Ran cloud radar through the night with stabalized platform on. Noted issues with absorbing foam and divider between the xmit and recieve dishes. Stoppled radar while at achnor in Hobart just after 00 UTC to adjust foam an inspect dishes. Removed big piece of duct tape that had fallen into one of the dishes. Restarted radar at ~01 UTC and ran until 0430 UTC approximately during which time the stabalized platfrom was not running. At 05 UTC Ian resarted the stabalized platform and I restarted the radar. All is running fine now.