|Subject:||CTD Altimeter calibration correction|
|When:||2015-12-02 08:24:12 UTC [entered into Elog as 2015-12-02 08:24:12]|
Position from underway logging
|Equipment:||CTD Linked to equipment register: CTD - Seabird 911 with 24 Bottle Rosette [view]|
The CTD calibration coefficients for the PA500 Altimeter were incorrectly set. a1 was -0.02442 instead of -0.01221. This has been corrected.
The correct values for this altimeter are:
a0 = 50
a1 = -0.01221
This error was evident on IN2015_V02 but not explained or corrected at the time. See the IN2015_V02 Instrumentation report for a plot of this.
|Logged by:||Ian McRobert [entered into Elog as Ian McRobert]|