|Subject:||not connected properly - air leak ino aerosol systems|
|When:|| 2021-06-13 18:55:07 UTC |
- entered into Elog/Everlog as 2021-06-13 18:55:07
Equipment: nano Scanning Mobility Particle Sizer (Grimm) [view]
Latitude: -19.09117 Longitude: 149.91117
- entered into Elog as 19 05.47 S, 149 54.67 E
16/6 23:26 - I just found that the air inlet tube was not connected properly. air has been sucked into the sampling system since I connected up the SMPS on 13/6 18:55. This will have affected the SMPS, and CPCs. It is possible that it will have affected the.nephelometer and CCNC.
It is possible, but I think unlikely, that it also affected the ozone and greenhouse gas system, depending on the interactions in the manifold and sample lines.